Does anyone have any information on how to calibrate a Hickok model tube tester? I’ve searched, but can’t seem to find any links. Thanks!. HAPPY HOLIDAYS TO ALL, Got a question about a Hickok I’ve got a chance to purchase one of these locally but in the picture I’m sending. After waiting years too long, I finally bought a really nice looking, lightly used Hickok and it was doing funny things with known good tubes. It read shorts until.
|Published (Last):||16 September 2009|
|PDF File Size:||19.40 Mb|
|ePub File Size:||9.18 Mb|
|Price:||Free* [*Free Regsitration Required]|
If you ever hiickok to see inside of a TV-7, it is really built. 5336 Sun 27, 2: I have other Hickok questions, so if there’s someone knowledgeable out there who wouldn’t mind talking to me on the phone, just send me an email with a number and a good time to call, and I’ll ring up, unless you happen to be in the Virginia Tidewater area and are willing to give a hands-on tutorial.
Does any one have a model that they could take a picture of the spool resistor stacks and email to me? Tubes Asylum Questions about tubes and gear that glows. How to test tubes accurately. Laid out something like a A, it has switch selected gm scales and a built in roll chart.
Topic – Good, minimum readings for Hickok ? It is classic military construction with potted transformers, ceramic sockets on the minis and octal, and hermetically sealed caps. Good, minimum readings for Hickok ? And the Hickok guy hickoi the is very different from the s or s. I can have your old meter rebuilt to proper specifications. I have the “spool assy removed” Torrance, CA Contact Email: Select a valid country. As Dave says, it sounds like somebody was playing “Nick, knack, paddywhack” inside there with the innards.
Thanks for your help! By submitting a comment, you are declaring that you agree with these rules:. This is due to the age and condition of the power transformers in these vintage 563. Get the item you ordered or get your money back. Back to home page. Yes, my password hlckok You guys gals too!
Dynamic Mutual Conductance Tube Tester 536
See other items More Delivery times may vary, especially during peak periods. Email to friends Share on Facebook – opens in a new window hikcok tab Share on Twitter – opens in a new window or tab Share on Pinterest – opens in a new window or tab. Sell now – Have one to sell? The spool resistors would hiickok been all right if they hadn’t used corrosive flux.
You must log in or sign up to reply here. Has switch selected micromho ranges.
The second question is hard to understand. Email to friends Share on Facebook 56 opens in a new window or tab Share on Twitter – opens in a new window or tab Share on Pinterest – opens in a new window or tab Add to watch list.
Do you already have an account? Mar Thu 27, 8: I test a lot of gickok octal tubes, and they are all on the chart!
This tube is measured with two laboratory testers to an accuracy of 1. This tester will give you hickokk most repeatable tests. If most of the screws are gone it means that somebody knowledgeable looked inside and decided the tester was SHOT and there was no point in bothering to put yickok the screws back in. Why Hickok went with off-the-wall non-standard panel meter specs I don’t know.
If you have a picture of the wiring of the spool resistor stack it would help lots.
Dynamic Mutual Conductance Tube Tester Equipment Hickok Elec
I would try to repair the old wooden spool resistor. A recommended list of Hickok tube testers. Models and A. Log in or Sign up. Dec Mon 31, This amount is subject to change until you make payment. I might be able to help. Hickok Electrical Instrument Company. Previous topic Next topic. These models are compact bench top models that are basically hikcok out as this A above.
I have no idea of these two models which is the better one. You can also visit my eBay store. The has sockets up to 9-pin mini.
They always break at the solder joint, it seems I’ve never encountered one before this.